Acetic Acid Detection Tubes 5/a (5-80 ppm)
Drager Short-term Tubes have proven to be a very cost-effective and reliable method for the measurement of gases.
For short-term measurements more than 160 Drager-Tubes are available to measure the so-called spot concentrations of specific gases. Measurement usually takes between 4 seconds and 15 minutes. The tubes are used either manually with the Drager-Tube pump accuro or automatically with the Drager X-act 5000.
A wide range of different gases and vapours can be detected and measured, e.g.: Determining and measuring concentration peaks, Measuring personal exposure in the inhalation area, Detecting leaks in pipelines, gas installations etc., and Air analysis in sewers, shafts and confined spaces.
|Standard Measuring Range:
||5 to 80 ppm
|Number of Strokes (n):
|Time for Measurement:
||approx. 30 sec.
||±10 to 15 %
||blue violet -> yellow
||10 to 40° C
||< 30mg H2O / L
||CH3COOH + pH Indicator ─> yellow reaction product
||It is impossible to measure acetic acid in the presence of other acids.
Organic acids are indicated by the same color change, but with different sensitivities.
Mineral acids (e.g. hydrochloric acid) are indicated by red discoloration and different sensitivities.